Imaging
Under institutional review board approval, 100 DCEMRI examinations of the liver from 35 patients were included in this study (women, 8; men, 27; age at examination, 51–83 years; number of examinations per patient, 1–4). The patients were imaged for about 3 minutes using a repeated breathhold paradigm (5). For this paradigm, the patients were instructed to initially hold their breath for as long as they could and then hold their breath repeatedly with a single deep inhalation in between each breath hold. Images acquired during deep inhalations showed large liver movement and severe motion blur. Therefore, these were excluded from further analysis. Dynamic MR imaging was started at the beginning of the first breath hold and a gadobenatedimegluminebased contrast agent (MultiHance, Bracco S.p.A., Italy) was administered intravenously.
A 3 T MRI scanner (Magnetom Skyra, Siemens Medical Systems, Erlangen, Germany) was used for imaging. A 3dimensional gradient echo sequence that speeds up acquisition using stochastic sampling with view sharing (timeresolved angiography with stochastic trajectories; TWIST) was used for 10 patients, and a sequence that enhances speed through partial Fourier reconstruction (volume interpolated breath hold examination; VIBE) was used for 25 patients. Sequence parameters varied among the patients (Table 1).
Table 1.
DCEMRI Sequence Settings for TWIST and VIBE Sequences

TWIST 
VIBE 
Flip angle 
10–20° 
9° 
Repetition time 
2.5–2.53 ms 
2.22 ms 
Echo time 
0.84–0.86 ms 
0.78–0.79 ms 
Temporal resolution 
2.3–4.3 s 
1.99–6.82 s 
Voxel volume 
15.6–45.8 mm^{3}

34.2–100 mm^{3}

Number of phases 
52–60 
38–60 
Acquisition time 
2.0–3.7 min 
2.0–6.7 min 
Number of examinations 
31 
69 
During a breath hold, the liver should ideally be still. Unfortunately, partial inhalation during a breath hold, imperfect reproducibility of the exhale state, and changes to patient posture result in liver movement between images. These movements, however, are smaller than those observed during free breathing. For the spatial resolution, signaltonoise ratio, and artifact level seen in this study, we, therefore, chose to use rigid registration to correct for the observed range of motions.
Consistent Solution
The registration of a pair (i, j) of phases i and j results in a rigid transform described by a rotation matrix Bi,j and a translation vector bi,j. Thus, given a structure found in phase i at spatial coordinates r(i)=(x, y, z)T, the corresponding structure, as indicated by the registration, can be found at position r(j)=Bi,jr(i)+bi,j in phase j.
Let us assume that there is a transform given by rotation matrix Ai and translation vector ai, which transforms the coordinates of any structure in the reference phase iref into the coordinates of the same structure in phase i. If all of the coordinates for the voxels inside the liver ROI are inserted into the columns of a matrix R, the corresponding points in phase j are given by AjR+aj1, where 1=[1, . . . , 1]1×nvoxels. However, if the registration of phase pair (i, j) is successful, the same coordinates are also given by the application of the true transform from the reference phase to phase i followed by the registration transform from i to j such that: such that:
We then have
npair of equations of the same form as equation
(1), 1 for each registered phase pair, and up to
nphases(nphases−1) are registered for a given stage. However, there are only
nphases−1 true transforms given by
Ai and
ai. Therefore, the system in equation
(1) is overdetermined if
npair>nphases−1. Because the registration of a phase pair may not produce the true transform but only an estimate, equation
(1) is not valid for all registered pairs
(i, j) in general. Therefore, we must choose in what sense to solve the system in equation
(1). If we define equation
(2) as follows:
we can observe that
‖Ei,j‖F2 is the sum of the squared distances between the liver voxel coordinates given by
Bi,j(AiR+ai1)+bi,j1 and
AjR+aj1. Finding the true transforms given by
Ai and
ai by minimizing the following equation:
where
‖⋅‖F is the Frobenius norm and would produce the least squares solution. If the error in the mapping of the liver voxel coordinates could be described by a normally distributed random variable, then minimizing equation
(3) would produce a good solution for
Ai and
ai. However, because some of the phase pair registrations could fail and produce severely erroneous transforms,
Bi,j and
bi,j, the sum in equation
(3) could be dominated by a few large terms that would thwart the accurate estimation of
Ai and
ai. Therefore, to lessen the impact of outlier registration transforms,
Ai and
ai can be robustly estimated by minimizing the following equation:
with
p<2 (
p=1/2 for this work). This can be done by using the iteratively reweighted least squares algorithm. For iteratively reweighted least squares, the following equation is solved repeatedly:
where
and
β is the set of all transform parameters
{Ai} and
{ai}, and
β(k) is the
k th iteration of
β. Equation
(5) is iterated until convergence. To avoid division by 0 in equation
(6), a nonzero regularization term
E0 is needed. As
‖Ei,j‖Fnpairs−1/2 is the root mean squared difference of the liver voxel coordinates produced by the lefthand side (LHS) and righthand side (RHS) of equation
(1),
E0npairs−1/2 should be smaller than the best expected registration accuracy. For this work,
E0npairs−1/2 was selected as 0.01 mm.
Each iteration of equation (5) amounts to solving a constrained, weightedleastsquares problem, where the constraint is that all Ai must be rotation matrices. Because the minimization problem in equation (5) is a rigidbody problem, it can be broken up into 2 independent problems, 1 problem for the translation of the liver center of mass and 1 problem for the rotation of the liver about its center of mass. This can be seen by noting that:
where
and
L is the lower triangular matrix in the Cholesky decomposition of
Σ=LLT. Thus, the problem in equation
(5) can be written as the following 2 problems:
where equation
(14) is a problem to find the rotation matrices
β1={Ai} only, whereas equation
(15) is a problem to find the transformed translation vectors
β2={a˜i} only and
β={β1, β2}. The problem in equation
(15) is a system of linear equations of the following form:
that can be solved in a least squares sense for
{a˜i} using the Moore–Penrose pseudoinverse. When solving this system, the solution from equation
(14) is used to fix
a˜iref to a constant by setting
airef=0. The problem in equation
(14) can similarly be written as a system of matrix equations as follows:
To find a solution for equation (17), a heuristic algorithm was used as described below. By summing over i, equation (17) can be rewritten as follows:
where the RHS is an approximation of the first term in the LHS, with an error given by the second term of the LHS. This formula can be solved recursively. However,
{Ai} is a rotation matrix and needs to be orthogonal with
det (Ai)=1. To ensure this, the RHS of equation
(18) can be decomposed into its singular value decomposition to produce a set of the following iteration equations:
where the sign of the last diagonal element in equation
(20) is selected to make the determinant of
Cj(l+1) positive and equation
(21) is needed to ensure that
Airef(l+1)=I. The stopping criteria for equations (
19) to (
21) are as follows:
Empirically, the outer recursive sequence given by equation (5) was observed to converge within 20 iterations for a few test cases. Therefore, 20 iterations of equation (5) were used for all examinations.
Evaluation
The effect of registration on the mean signal intensity versus time for a 1cm^{3} cubic ROI placed 1 cm below the dome of the liver for 1 patient is illustrated in Figure 2. The spikes seen in the intensity for the nonregistered time series are caused by tissue moving in and out of the ROI due to breathing.
Figure 2.
Signal intensity from an intrahepatic region of interest (ROI) from a dynamic contrastenhanced magnetic resonance imaging (DCEMRI) examination. The left side of the plot shows the time–intensity curves for a 1cm^{3} cube placed 1 cm below the dome of the liver of a patient. The black curve shows the intensity before registration, and the gray curve shows the intensity after robust registration. Phases corresponding to deep inhalations during the examination are shown as circles and are not included in the main curves. The right box illustrates the local mean signal intensity (red line) and the local spike area (solid red triangle) around a sample (red dot).
Given the appearance of the signal variations seen in Figure 2, a suitable metric for the quality of the DCEMRI time series should reflect how smooth the time–intensity curves are for the voxels inside the liver ROI. The temporal resolution of the time series as compared with the rise time of the time–intensity curve should also be reflected by the alignment quality metric. If only a few samples are collected during the rise time of a curve, the impact of 1 erroneous sample will be greater than that if a larger number of samples are collected. The variation of temporal sampling density across examinations can be taken into account by transforming the time of a given phase t into a normalized time τ as follows:
where
t0 is the time at the center of the intensity ramp and
T is half of the rise time. Thus, the rise time of the mean liver intensity for each examination is similar after transformation. The values of
t0 and
T were determined by fitting the time–intensity curve for the whole liver for each examination to the phenomenological function as follows:
where
0 is the baseline intensity,
Δmax is the maximum signal enhancement, and
α is the decay rate of the plateau. The function was fitted to the data using minimization of the
L1norm of the signal residuals.
Furthermore, errors in the baseline will be more troublesome for parameter estimation than had they appeared in the postcontrast plateau. In light of these considerations, we have chosen to use an approximation of the local spike area divided by the mean temporally local signal intensity to calculate the metric (see right panel of Figure 2 for illustration).
If the normalized temporal spacing between phases is Δτ, an approximation of the timenormalized, intensitynormalized spike area (TISA) metric, m(r, i), at phase i and position r is then given by the following equation:
where the numerator is the spike area and the denominator is the mean signal intensity of the voxel for the phase and its 2 temporal neighbors. Because the preceding and the following phases are needed to evaluate equation
(25) for a phase, equation
(25) cannot be evaluated for phases adjacent to others omitted because of deep inhalations during the examination.
Research Articles
Download PDF (796.08 KB)
TOMOGRAPHY, September 2016, Volume 2, Issue 3:188196
DOI: 10.18383/j.tom.2016.00145
An Overdetermined System of Transform Equations in Support of Robust DCEMRI Registration with Outlier Rejection
Adam Johansson^{1}, James Balter^{1}, Mary Feng^{1}, Yue Cao^{1}
Abstract
Quantitative hepatic perfusion parameters derived by fitting dynamic contrastenhanced (DCE) magnetic resonance imaging (MRI) of liver to a pharmacokinetic model are prone to errors if the dynamic images are not corrected for respiratory motion by image registration. The contrastinduced intensity variations in pre and postcontrast phases pose challenges for the accuracy of image registration. We propose an overdetermined system of transformation equations between the image volumes in the DCEMRI series to achieve robust alignment. In this method, we register each volume to every other volume. From the transforms produced by all pairwise registrations, we constructed an overdetermined system of transform equations that was solved robustly by minimizing the L_{1/2}norm of the residuals. This method was evaluated on a set of 100 liver DCEMRI examinations from 35 patients by examining the area under spikes appearing in the voxel time–intensity curves. The robust alignment procedure significantly reduced the area under intensity spikes compared with unregistered volumes (P < .001) and volumes registered to a single reference phase (P < .001). Our registration procedure provides a larger number of reliable time–intensity curve samples. The additional reliable samples in the precontrast baseline are important for calculating the postcontrast signal enhancement and thereby for converting intensity to contrast concentration. On the intensity ramp, retained samples help to better describe the uptake dynamics, providing a better foundation for parameter estimation. The presented method also simplifies the analysis of data sets with many patients by eliminating the need for manual intervention during registration.
Introduction
Dynamic contrastenhanced magnetic resonance imaging (DCEMRI) in the liver enables quantification of hepatic perfusion for assessment of tumor response to radiation therapy, as well as normal tissue damage (1–5). One of the challenges in liver DCEMRI is respiratory motion during acquisition, which can cause blurring and artifacts in the images and intensity variations in the dynamic curves, and subsequently, errors in perfusion quantification using pharmacokinetic (PK) models. To reduce respiratory motioninduced degradation of liver images during DCEMRI acquisition, a breathing control paradigm has been proposed, during which the patient is instructed to hold their breath multiple times (taking breaths in between) (5). The dynamic image volumes acquired during the 2–3 minutes of contrast uptake are still misaligned because of poor reproducibility of the liver position during multiple breath holds, loss of control during breath holding, and occasionally gross movement of the patient. Therefore, a voxel cannot be assumed to cover the same piece of tissue throughout the whole examination, but, instead, it covers different pieces at different points in time. Therefore, if not corrected for, breathing and gastrointestinal motions can induce artifacts, such as intensity spikes and wide bumps, in the contrast uptake curves. To estimate the arterial and portal venous perfusion in the liver, a sufficient number of reliable samples of the uptake curve must be collected to fully describe the contrast uptake dynamics. The shape of the precontrast plateau and the beginning of the intensity ramp following contrast agent administration can have a large effect on the estimated perfusion parameters, in particular on hepatic arterial perfusion. Unwanted artifacts introduced into the uptake curve by patient motion reduce the number of reliable samples, increasing the risk of inaccurate PK model parameter estimation.
To correct for image misalignment in the DCE series, image registration has been used to align the magnetic resonance (MR) images at different points in time or phases (1, 6–8). However, registration of the DCEMRI phases has some unique challenges, for example, a large number of dynamic image volumes, low spatial resolution, and low contrasttonoise ratio that are tradeoffs for the high temporal resolution. In addition, the image contrast is changed completely over the whole course from precontrast to rapid contrast uptake and to full contrast enhancement. Combined with all these effects, the dynamic contrast agentinduced intensity changes pose a difficult problem even to registration methods using multimodality imagetoimage metrics such as mutual information (9–11).
A number of methods have been developed to tackle the DCEMRI registration problem. Some methods attempt to register each phase to a reference phase selected among all phases. Such methods use either regionlimited rigid registration (12) or deformable registration (9, 10). Deformable registration may yield incorrect volume changes around contrastenhanced lesions unless the method is adapted to explicitly counteract such changes (11). As an alternative to reference phasebased registration, image intensity changes can be modeled using a PK model and the model incorporated into the registration algorithm (13–16). Such PKbased methods can be slow because of the need for repeated PK model parameter estimation. For a faster procedure, principal component analysis (PCA) can be used to generate reference images for registration (17). However, PCA may not be able to adequately describe the spatially varying time–intensity curves. Therefore, the phases could alternatively be decomposed into a lowrank and a sparse image component in a procedure called robust PCA that can be combined with image registration (18, 19). Robust PCA can also be used to combine registration with reconstruction, but this requires access to raw data from the MRI scanner (20). However, when comparing the 3 representative registration methods (21), 1 PK modelbased, 1 PCAbased, and 1 sequentially registering each phase to its preceding phase, the sequential registration method performed better (average error 14.7% in estimated parameters) than the more complex iterative, PCAbased and PK modelbased methods (average errors: 39.5% and 39.2%, respectively). Nevertheless, all image registration methods reduced errors in the estimated parameters in the tumors in comparison with without image registration methods.
The performance of the sequential method can be understood by considering that the changes between neighboring phases are small. Therefore, registration has a greater chance of success. However, misregistration errors are accumulated and propagated from the earlier phases to the later phases. If registration at the early phase fails, all following phases will be incorrectly aligned with the first phase.
Inspired by the findings in Rajaraman S et al.'s study (21), and taking into account the aforementioned drawback of the sequential registration procedure, we introduce an overdetermined system for achieving robust registration for the DCE image series. In this procedure, rather than registering each phase to a reference phase or to a preceding phase, each phase is registered to every other phase. This produces an overdetermined system of transform equations that can be solved using robust statistical methods to reject outliers corresponding to failed registrations. This new method is evaluated on a set of 100 liver DCEMRI examinations, and the results are compared with those of a conventional registration method where all phases are registered to 1 postcontrast reference phase. A robust registration method for liver DCEMRI would greatly simplify the workflow of clinical liver perfusion studies with a large number of participating patients by eliminating the need for manual intervention following faulty registration.
Materials and Methods
Imaging
Under institutional review board approval, 100 DCEMRI examinations of the liver from 35 patients were included in this study (women, 8; men, 27; age at examination, 51–83 years; number of examinations per patient, 1–4). The patients were imaged for about 3 minutes using a repeated breathhold paradigm (5). For this paradigm, the patients were instructed to initially hold their breath for as long as they could and then hold their breath repeatedly with a single deep inhalation in between each breath hold. Images acquired during deep inhalations showed large liver movement and severe motion blur. Therefore, these were excluded from further analysis. Dynamic MR imaging was started at the beginning of the first breath hold and a gadobenatedimegluminebased contrast agent (MultiHance, Bracco S.p.A., Italy) was administered intravenously.
A 3 T MRI scanner (Magnetom Skyra, Siemens Medical Systems, Erlangen, Germany) was used for imaging. A 3dimensional gradient echo sequence that speeds up acquisition using stochastic sampling with view sharing (timeresolved angiography with stochastic trajectories; TWIST) was used for 10 patients, and a sequence that enhances speed through partial Fourier reconstruction (volume interpolated breath hold examination; VIBE) was used for 25 patients. Sequence parameters varied among the patients (Table 1).
Table 1.
DCEMRI Sequence Settings for TWIST and VIBE Sequences
i] Abbreviations: DCEMRI, Dynamic contrastenhanced magnetic resonance imaging; TWIST, timeresolved angiography with stochastic trajectories; VIBE, volume interpolated breath hold examination.
During a breath hold, the liver should ideally be still. Unfortunately, partial inhalation during a breath hold, imperfect reproducibility of the exhale state, and changes to patient posture result in liver movement between images. These movements, however, are smaller than those observed during free breathing. For the spatial resolution, signaltonoise ratio, and artifact level seen in this study, we, therefore, chose to use rigid registration to correct for the observed range of motions.
Preprocessing
Before registration, a number of preprocessing steps were performed. These were the same for the robust registration procedure (section Robust Registration) and the conventional procedure (section Conventional Registration) to which it was compared.
Reference Phase Selection.
The conventional procedure involved alignment of all phases to 1 reference phase. For each examination, this reference phase was selected among all phases as the phase with the smallest sum of mean absolute distances to the other phases in intensity, which is calculated as follows:
RegionLimited Image Registration.
Because we wish to correct for the motion of the liver with a rigid transform and the rest of the body inside the field of view (FOV) does not move in the same way as the liver, regionlimited registration was used. For this purpose, a region of interest (ROI) was drawn around the liver on the reference phase. This ROI was drawn snugly along the edge of the liver. For the regionlimited registration, it is advantageous to include a margin outside the liver ROI. This margin covers the contrastrich interface between the liver and the surrounding tissue, and it helps to drive the registration algorithm. The margin was set to 15 mm for all registrations.
Bias Field Correction.
MR images commonly contain a spatial intensity variation that is not the result of intrinsic tissue properties but of the position of the tissue with respect to the transmission and receiver radiofrequency coils. This “bias field” can cause misregistration if it is the dominant feature in an image. Fortunately, the bias field typically varies slowly with respect to the spatial position. The effect of the bias field on the images used for registration was reduced by dividing each phase by a filtered version of itself. The modified phases used for registration were then given by the following equation:
Robust Registration
The robust registration procedure for each DCE time series was divided into 5 stages—3 translational registration stages and 2 rigid registration stages with both translation and rotation. For the robust procedure, the workflow for each stage is illustrated in Figure 1. Either each possible pair of phases or each pair in a subset of all possible pairs was registered. Thus, each registered pair gave rise to a rigid transform describing the relative position of the 2 phases of the pair. For the registration of each pair, the first phase of the pair was used as a fixed image and the second as a moving image. The dilated liver ROI was applied as a mask to the fixed image to restrict the domain for which the registration metric was evaluated. Because of this asymmetry, the registration of pair( i , j ) , where i is the index of the fixed image and j is the index of the moving image, did not produce the same transform as the registration of pair ( j , i ) .
Figure 1.
Registration of each possible pair of phases. The transforms produced by each registration transform any coordinate in the fixed image coordinate system into its corresponding point in the moving image coordinate system.
After the registration of all pairs of 1 of the steps, the resulting transforms were used to produce a set of consistent transforms as described in the next section. These transforms were then applied to the phases before proceeding to the next stage.
The registration of the 2 phases of a pair was performed using the open source registration software Plastimatch, version 1.6.0beta (http://plastimatch.org). The settings of the translational and rigid registration steps are shown in Table 2. For the robust procedure, the first 2 translational steps only registered a subset with10 n phases pairs selected randomly among the n phases ( n phases − 1 ) possible pairs. These 2 steps helped roughly align the liver ROI for all the phases in preparation for the full registrations performed by the third through the fifth stage.
Table 2.
Plastimatch Parameters for Translational and Rigid Registration
Consistent Solution
The registration of a pair( i , j ) of phases i and j results in a rigid transform described by a rotation matrix B i , j and a translation vector b i , j . Thus, given a structure found in phase i at spatial coordinates r ( i ) = ( x , y , z ) T , the corresponding structure, as indicated by the registration, can be found at position r ( j ) = B i , j r ( i ) + b i , j in phase j.
Let us assume that there is a transform given by rotation matrixA i and translation vector a i , which transforms the coordinates of any structure in the reference phase i ref into the coordinates of the same structure in phase i . If all of the coordinates for the voxels inside the liver ROI are inserted into the columns of a matrix R , the corresponding points in phase j are given by A j R + a j 1 , where 1 = [ 1 , . . . , 1 ] 1 × n voxels . However, if the registration of phase pair ( i , j ) is successful, the same coordinates are also given by the application of the true transform from the reference phase to phase i followed by the registration transform from i to j such that: such that:
(1)
(2)
(3)
(4)
(5)
(6)
Each iteration of equation (5) amounts to solving a constrained, weightedleastsquares problem, where the constraint is that allA i must be rotation matrices. Because the minimization problem in equation (5) is a rigidbody problem, it can be broken up into 2 independent problems, 1 problem for the translation of the liver center of mass and 1 problem for the rotation of the liver about its center of mass. This can be seen by noting that:
(7)
(8)
(9)
(10)
(11)
(12)
(13)
(14)
(15)
(16)
(17)
To find a solution for equation (17), a heuristic algorithm was used as described below. By summing overi , equation (17) can be rewritten as follows:
(18)
(19)
(20)
(21)
(22)
Empirically, the outer recursive sequence given by equation (5) was observed to converge within 20 iterations for a few test cases. Therefore, 20 iterations of equation (5) were used for all examinations.
Conventional Registration
A conventional registration procedure was used as a benchmark for the robust registration. This conventional procedure registered each phase to the reference phase using the same 5 registration stages as for the robust procedure. However, phases were not registered to anything but the reference phase, and no processing of the resulting transforms was performed except the combination of the transforms from the 5 stages for each phase. The same ROI was used to limit the registration region, and the settings in Table 2 were used for alignment.
Evaluation
The effect of registration on the mean signal intensity versus time for a 1cm^{3} cubic ROI placed 1 cm below the dome of the liver for 1 patient is illustrated in Figure 2. The spikes seen in the intensity for the nonregistered time series are caused by tissue moving in and out of the ROI due to breathing.
Figure 2.
Signal intensity from an intrahepatic region of interest (ROI) from a dynamic contrastenhanced magnetic resonance imaging (DCEMRI) examination. The left side of the plot shows the time–intensity curves for a 1cm^{3} cube placed 1 cm below the dome of the liver of a patient. The black curve shows the intensity before registration, and the gray curve shows the intensity after robust registration. Phases corresponding to deep inhalations during the examination are shown as circles and are not included in the main curves. The right box illustrates the local mean signal intensity (red line) and the local spike area (solid red triangle) around a sample (red dot).
Given the appearance of the signal variations seen in Figure 2, a suitable metric for the quality of the DCEMRI time series should reflect how smooth the time–intensity curves are for the voxels inside the liver ROI. The temporal resolution of the time series as compared with the rise time of the time–intensity curve should also be reflected by the alignment quality metric. If only a few samples are collected during the rise time of a curve, the impact of 1 erroneous sample will be greater than that if a larger number of samples are collected. The variation of temporal sampling density across examinations can be taken into account by transforming the time of a given phaset into a normalized time τ as follows:
(23)
(24)
Furthermore, errors in the baseline will be more troublesome for parameter estimation than had they appeared in the postcontrast plateau. In light of these considerations, we have chosen to use an approximation of the local spike area divided by the mean temporally local signal intensity to calculate the metric (see right panel of Figure 2 for illustration).
If the normalized temporal spacing between phases isΔ τ , an approximation of the timenormalized, intensitynormalized spike area (TISA) metric, m ( r , i ) , at phase i and position r is then given by the following equation:
(25)
Results
After the consistent registration solution has been found from the overdetermined system of transform equations by minimizing equation (4), the individual norms‖ E i , j ‖ F show how well a particular transform { B i , j , b i , j } agrees with the overall solution. This is illustrated in Figure 3 for the last registration stage of 1 examination. The small dark square in the lower left corner of the figure represents the registration errors among the precontrast phases. The larger dark square in the upper right corner represent the errors among the postcontrast phases. These 2 groups have relatively small deviations from the consistent solution compared with the errors found for registrations between pre and postcontrast phases as represented by the brighter rectangles in the upper left and lower right corners. This pattern was seen for a majority of the examinations registered and indicates that registrations between pre and postcontrast phases are more likely to produce transforms that do not agree with the consistent solution.
Figure 3.
The value of‖ E i , j ‖ F in millimeters for 1 examination following minimization of equation (4) is indicated by the color in the figure. The Xaxis shows the phase index of the fixed registration phase and the Yaxis shows the index of the moving registration phase. Nine phases corresponding to deep inhalations have been omitted from the figure. Phase index 16 corresponds to the middle of the intensity ramp (τ = 0 ). Indices <16 are precontrast phases and those >16 are postcontrast phases.
By sampling the intensity along a 1×1cmthick column extending 20 cm above and 20 cm below the liver dome, the effect of registration on the position of the liver can be illustrated. Figure 4 shows the intensity along such a column for the phases of 1 examination along with the time–intensity curves for a 1cm^{3} cubic ROI placed 1 cm below the liver dome. The inferosuperior intensity profile exhibits motion before registration. After conventional registration, the motion is reduced for most phases, but it fails for a few precontrast phases. The intensity profile produced by the robust registration method has no such obviously failed registrations. Failed registrations like the one seen in Figure 4C–D were not found among any of the registrations produced by the robust method.
Figure 4.
The intensity along a 40cmlong column extending in the inferosuperior direction is seen for the phases before registration (A), after conventional registration (C), and after robust registration (E). The mean intensity of a 1cm^{3} cubic ROI placed 1 cm below the liver dome is seen as a solid black curve for the phases before registration (B), after conventional registration (D), and after robust registration (F). The black circles indicate the intensities of phases excluded because of deep inhalations during examination. For each column of (A), (C), and (E), the intensity profile was divided by its mean to improve the visual contrast of the figure.
The mean TISA,m , for all examinations is shown with respect to the normalized time in Figure 5B. As a reference to help interpret the normalized time, τ , the mean of all liver time–intensity curves is shown in Figure 5A. This figure further illustrates where a particular value of τ corresponds to the precontrast region (τ < − 1.5 ), the intensity ramp (− 1.5 < τ < 1.5 ) , or the postcontrast region ( t > 1.5 ). The TISA curves are consistently higher in the precontrast region compared with those in the postcontrast region. This is a result of the lower mean intensity of the phases in the precontrast region. In addition, at the beginning of the intensity ramp (around τ = − 1 ) , the conventional registration procedure performs almost as poorly as no registration, as seen by the close proximity of their TISA curves (green and red). Confidence intervals for the paired TISA differences between the robust method, the conventional method, and no registration are given in Table 3. The robust method is significantly better than no registration for the whole time interval (paired t test, P < .001). It is also significantly better than the conventional method for the precontrast and early contrast regions (P < .001).
Figure 5.
The mean time–intensity curve from all examinations is shown in (A). The mean timenormalized intensitynormalized spike area,m , is shown in (B). For (B), the mean of m was calculated for all examinations and a time window that was one normalized time unit wide. The colored bands show the 95% confidence interval for the mean.
Table 3.
Paired TISA Difference for 3 Registration Methods and Time Intervals
i] Abbreviations: TISA, timenormalized, intensitynormalized spike area; SD, standard deviation; CI, confidence interval.
Discussion
In this study, we have implemented an overdetermined registration procedure for liver DCEMRI examinations. We found that our procedure significantly reduced the image registration errors in the DCE time curves, particularly in the precontrast and early contrast uptake phases, compared with a conventional referencephasebased registration method. Registration errors in the precontrast baseline could profoundly affect the calculation of signal intensity changes and the conversion to contrast concentration, whereas errors in the early contrast uptake curve propagate into derived parameters that are sensitive to the early curve dynamics, for example, hepatic arterial perfusion in liver DCEMRI. The improved DCE curves describe the contrast agent uptake dynamics more correctly, and thereby, provide a better foundation for PK model parameter estimation.
DCE images acquired at different parts of the dynamic curves have different sensitivities to image registration errors. We found that registering precontrast phases to postcontrast phases is more likely to result in registration errors than registering within the precontrast phases or within the postcontrast phases. The conventional referencebased registration method likely results in a higher misregistration rate for the precontrast phases, as well as the early contrast uptake phases, if the postcontrast phase is chosen as reference. As an effect of this phenomenon, spikes can appear in or around the baseline plateau in the time–intensity curves. The parameters of a PK model may still be possible to estimate with such spikes present by using robust regression methods. However, each registration failure reduces the number of reliable samples. Even worse, if sample points are lost on the intensity ramp by removing the misregistration points, the remaining samples may not be able to correctly describe the dynamics of the curve, resulting in faulty PK parameter estimates. Some parameters are highly sensitive to the shape of the initial ramp of the uptake curve, for instance, hepatic arterial perfusion in the liver. The robust registration procedure presented reduces the number of failed registrations in the precontrast region by rejecting them as outliers in an overdetermined system of transform equations and thereby provides a greater number of reliable samples for parameter estimation.
For the postcontrast plateau, the robust registration method and the conventional registration method perform almost equally well. This is a result of the higher signaltonoise ratio and more stable contrast for the phases of this region. Both registration methods produce smoother curves than no registration. The robust method is seen to have a slightly lower TISA than the conventional method. This is because the robust method takes the mutual information of all pairs of phases into account when estimating the transform. Therefore, the robust method acts as a more efficient estimator for the true transforms than does the conventional method.
In this work, we performed each registration separately and then combined the resulting transforms using a method that rejects outliers. Another possible approach would be to maximize the sum of the mutual information metrics of all pairs of phases using strategies similar to those reported by others (22–24). Such an approach might improve the registration of DCEMRI further. In addition, although the method presented in this paper is only applicable to rigid registration, a method based on the maximization of pairwise mutual information could aid in improving the robustness of deformable registration for DCEMRI. Robust deformable registration could allow free breathing examinations, which would relieve subjects from the burden of repeatedly holding their breath.
The robust method produced a consistent registration quality across all examinations, without failed alignments such as those observed for the conventional method. It therefore reduces the need for manual intervention following registration and helps to simplify the workflow of clinical studies with many subjects.
Conclusions
A method has been presented that registers every possible pair of phases from a DCEMRI examination and derives a final set of transforms from them usingL 1 / 2 minimization. The method is robust, insofar as to be able to reject such failed registrations that can appear when registering precontrast to postcontrast phases. The robust registration method improves the smoothness of the resulting time–intensity curves of voxels in the liver by eliminating spurious intensity spikes induced by motion or failed registration. Therefore, a baseline with reduced bias and a larger number of reliable samples in the intensity ramp are made available for the PK model parameter estimation. The consistent registration quality produced for all examinations studied shows that the method could simplify the workflow of clinical studies with many patients by eliminating the need for manual intervention following registration.
Notes
[3] Abbreviations:
DCE
Dynamic contrastenhanced
PK
pharmacokinetic
MR
magnetic resonance
PCA
principal component analysis
ROI
region of interest
RHS
righthand side
LHS
lefthand side
TISA
timenormalized, intensitynormalized spike area
Acknowledgments
This work is supported, in part, by NIH/NCI RO1 CA132834 and PO1 CA059827.
Disclosure: No disclosures to report.
References
Journal Information
Journal ID (nlmta): tom
Journal ID (publisherid): TOMOG
Title: Tomography
Subtitle: A Journal for Imaging Research
Abbreviated Title: Tomography
ISSN (print): 23791381
ISSN (electronic): 2379139X
Publisher: Grapho Publications, LLC (Ann Abor, Michigan)
Article Information
Copyright statement: © 2016 The Authors. Published by Grapho Publications, LLC
Copyright: 2016, Grapho Publications, LLC
License (openaccess, http://creativecommons.org/licenses/byncnd/4.0/):
This is an open access article under the CC BYNCND license (http://creativecommons.org/licenses/byncnd/4.0/).
Publication date (print): September 2016
Volume: 2
Issue: 3
Pages: 188196
Publisher ID: TOM0014516
DOI: 10.18383/j.tom.2016.00145
PDF
Download the article PDF (796.08 KB)
Download the full issue PDF (190.64 MB)
Mobileready Flipbook
View the full issue as a flipbook (Desktop and Mobileready)